In this video, Dr. Nasseh shares some of the specific memory settings as it relates to safety and efficiency and makes some clinical recommendations for specific settings
TOOL USED IN VIDEO
ENDOSYNC A.I.1P APEX LOCATOR
This topic contains 2 replies, has 2 voices, and was last updated by Allen Ali Nasseh 3 weeks, 1 day ago.
Is there a way to run Wave one gold to auto reciprocate? Or just setting torque to 0.2?
@Jeffthedentist: yes, you can do the same thing with WOG. But you need to use EndoSync Plus, not the original EndoSync as you will need Reverse OTR with 0.2Ncm setting with a cutting angle of 150 and non-cutting angle of 30.
Hope this helps.
You must be logged in to reply to this topic.
EndoSyncPlus Overview Part 1 of 3
EndoSyncPlus Overview Part 2 of 3.
Single Stroke & Clean (SSC) vs. OTR vs. Rhythm Motion! (Friday Questions!)
EndoSync Handpiece and EndoSync A.I. Apex Locator Review
Apical Stop Work Around for EndoSync / EndoSync A.I. (Quick Tutorials)